When using appropriate accessories the ultrasound application will be improved and the oscillating tank and parts to be cleaned will simultaneously be protected.
During the ultrasonic cleaning process the parts to be cleaned or vessels must not be placed on the tank bottom, as it could provoke abrasion between the parts and the tank itself due to the vibration of the transducers . Therefore it is recommended to use an appropriate cleaning basket. The BANDELIN cleaning baskets are constructed in that manner that the parts to be cleaned are always in the optimum range of cavitation ( 1-2 cm from the bottom ) during the ultrasonic process. Using a basket allows a comfortable placement of the parts to be cleaned in the bath and removal after cleaning, to rinse the cleaned part, without getting in contact with the cleaning or disinfecting liquid.
Further information is available in the table of measurements for the insert baskets.
Suitable insert baskets can be found in the data sheet of the ultrasonic bath.
A suitable lid for an ultrasonic bath protects the cleaning liquid from contamination and prevents odors from the cleaning or disinfecting liquid . Moreover, possibe annoying perceived noise emission during the operation of the ultrasonic cleaner is reduced. The lids are designed that way that any condensation is directed back into the ultrasonic bath .
Matching lids can be found in the data sheet of the ultrasonic bath.
Insert beakers and tubs for the indirect ultrasonic cleaning.
In case of using a cleaner or solvent reacting aggressive to stainless steel (eg TICKOPUR J 80 U), suitable plastic insert tubs or insert beakers made of glass or plastic are required to protect the cleaning unit. As the beakers must not be placed on the tank bottom, BANDELIN offers matching positioning lids for holding the beakers. A further advantage of using several beakers in an ultrasonic device is the possibility of the coexistence of different cleaning and disinfecting solutions without changing the bath. (eg removal of dental cements with STAMMOPUR Z and disinfection and cleaning of instruments with STAMMOPUR DR 8 in the dental practice)
Please note that for the transmission of the ultrasound to the tub/beakers the surface tension of water must be reduced and therefore a cleaning solution (eg TICKOPUR R 33-3%) should be used as a coupling liquid / contact liquid.
Holder ZF 10 / ZF 28 for laboratory flasks as fixing aid
In order to ensure proper sample preparation, appropriate prerequisites for laboratory work must be used.
Samples are to be homogenised, extracted or degassed quickly and reliably for subsequent analysis in laboratory vessels of various shapes and sizes. Fastening is often a special challenge. For this reason, BANDELIN has expanded the product range with new accessories: practical holders for laboratory flasks for fixing them into insert baskets for ultrasonic baths. Without effort and additional tools, the holder for laboratory flasks can be precisely fitted so that the sample preparation can be made place without problems. In addition to the previously known spring clamp EK, BANDELIN now offers an alternative for the basket sizes K 10/K 10 B and K 28/K 28 C:
ZF 10 for basket K 10/K 10 B – 5 holders for laboratory flasks 155 mm each and 3 holders for laboratory flasks 215 mm each
ZF 28 for basket K 28/K 28 C – 9 holders for laboratory flasks 215 mm each and 4 holders for laboratory flasks 420 mm each
The holders for laboratory flasks are available in three different lengths and can be mounted as a longitudinal and cross divider quickly and simply by means of small hooks at every position of the edge of the basket border. The user can thus determine the size individually depending on the flask shape and size (10 - 250 ml) and does not need any further tools. The stability of the lab flasks is guaranteed in this way. In no time, the division size can be changed by hand. A suitable connection to the contact liquid and thus loss-free ultrasound transmission into the sample to be treated is ensured.
In order to further ensure the 2/3-level in the SONOREX ultrasonic bath and to avoid flooding the laboratory flasks, we recommend the GV 10 handle adjustment.
SONOREX TECHNIK Peripherals
For the expansion of the industrial ultrasonic baths from the SONOREX TECHNIK program we offer suitable peripherals such as lifting devices , filtration units and oil separators as accessories.